Operating Instructions. Valid: These operating instructions are protected by copyright. Rights resulting the- reof, particularly reprint, photomechanical or digital postprocessing or reproduc- tion, even in part, are only allowed with the written consent of Thermo Electron LED GmbH. This regulation does not apply to reproductions for in-plant use. The contents of the operating instructions are subject to change without further notice.
|Country:||Turks & Caicos Islands|
|Published (Last):||15 April 2006|
|PDF File Size:||7.32 Mb|
|ePub File Size:||13.55 Mb|
|Price:||Free* [*Free Regsitration Required]|
Quick Links. Operating Instructions. General Information Table of Contents. Page 2 These operating instructions are protected by copyright. Rights resulting the- reof, particularly reprint, photomechanical or digital postprocessing or reproduc- tion, even in part, are only allowed with the written consent of Kendro Laboratory Products GmbH.
This regulation does not apply to reproductions for in-plant use. General notes NOTE is used for applicational hints and useful information. Refill ml water for ContraCon disinfection routine If CO is released, initiate safety measures immediately! High concentrations of nitrogen reduce the oxygen content in the air.
N2 released in large amounts into the room atmosphere may cause suffocation due to oxygen deficiency. If damages are detected or if components are missing, please contact the carrier and Kendro Laboratory Products immediately. Page Space Requirements The height of the floor stand should not fall below mm.
Several floor stands and carriers are available as options from Kendro part numbers see Section NOTE — Accessibility of the devices Tray Fig. Page Contracon Decontamination Routine The effectiveness of the ContraCon decontamination routine has been tested and certified by independent institutes. Information about these tests is available at request from Kendro Laboratory Products.
The entire program run of the ContraCon decontamination routine will take approx 25 hours. Page Fig. Gas connection: The gas supply line between the device and the gas supply system is connected using the supplied connecting hoses. For this purpose, a potential-free alarm contact  is preinstalled in the device.
A pressure compensation opening with insert  at the rear panel of the device allows a compensation between the pres- sures in the work space and in the operating room. Before the initial start-up, the device must be decontaminated. Make sure that the two vertical elements  of the shelf support are flush with the support rail. The operating pressure of the gas applied to the device must not exceed 1 bar.
Connect gas pressure hose  to the sleeve of the gas sup- ply system. Contact with current-carrying components may cause a lethal electric shock.
Before connecting the device to the power supply, check plug and connection line for damage. Do not use damaged components when con- necting the device to the power supply! The device must be connected only to a correctly installed and grounded power supply source Connection of the device: 1.
Turn PC off. Each integer value represents a current failure code in the assigned control loop incubati- on temperature, CO content, decontamination temperature, general, O con The connection of the device to an external The values for the operating voltage of the external circuits and of the fusing of the alarm system are given in the table below.
Turn the device on Press power switch. All indicators on the operating panel come on, all displays show the numeric value 8 to indicate that the test routine is being run. Enter the nominal value: Depending on the requirements for the working process, two different O con- trol ranges are preprogrammed at the factory.
Accept and store the new control state: Release both keys. The oxygen supply system is now disabled. Enabling the O control The O control can be reenabled at any time during the operation.
Indicate the nominal value: Press the key. The CO display shows the current nominal value. Activate the configuration mode: Keep the key depressed for 5 seconds, then release key. All indicators on the operating panel flash. Display the mode: Press the key. The temperature display shows the current mode high humidity. After the start, the device control adjusts the temperature to the set nominal value while humidity is generated.
Cancel the auto-start routine: Keep the depressed for 5 seconds. The displays returns to normal operation incubation operation. This system recogni- zes failures during the operation and allows the allocation of failure causes due to numeric codes.
Failure recognition is displayed by an acoustic and a visual alarm at the operating panel. The audible alarm sounds. Silence the audible alarm: Press any key. The audible alarm is silenced. Read failure codes: Keep the key depressed. If no failure is detected, each display shows three hyphens. If the system detects a failure, the display that is assigned to the corre- sponding control circuit shows a failure code.
Scroll through the failure code table: To read stored failure codes from the failure code table: Keep the key depressed. To scroll, keep the key depressed. The temperature display shows the 10 failure codes that had been regi- stered last. Turn the device off: Press the power switch.
All indicators go off. Turn the device back on: Press the power switch. Fill the water tray with up to max. Do not exceed the upper level mark. Make sure that the CO supply system valves are open.
If the work space surfaces are contaminated, germs my be transferred to the environment of the device. In case of a shut-down, the device must be decontaminated! ContraCon decontamination routine: is used to decontaminate the entire work space including all installed components and sensors in an automated program run. Page 59 In these areas, some other suited disinfectant that meets the aforementioned safety requirements must be used.
As an alternative for the U. Direct order: www. Prior to cleaning and disinfection work, discon- nect the device from the power supply! Remove all samples from the work space and store them at a safe place. Spray disinfectant onto the surfaces of the work space and of the accessories or wipe the surfaces clean using disinfectant.
Again, spray disinfectant onto the surfaces of the work space and of the ac- cessories or wipe the surfaces clean using disinfectant. Each individual phase or se- veral phases can be cancelled i.
The ContraCon de- contamination routine must then be ended by pressing the appropriate key. Turn the device on: Press the power switch. All indicators at the control panel illuminate. The software version is shown at the temperature display and at the CO display. Page Maintenance Refill the work space water tray with fresh processed water.
NOTE — Service contract Kendro Laboratory Products offer a device- specific service contact that comprises all test and service works required. If a major temperature deviation is found during this check, a temperature calibration is required. Activate calibration: Keep the key depressed for 5 seconds. All operating panel indicators flash. Display the nominal value: Press the key. If a major deviation is found during this check, a CO calibration is required.
Cancel the calibration process: Press any key. The temperature display and the CO display show the actual values. No tools are required to replace the seal. Page Spare Parts And Accessories When ordering spare parts, please have the device specifications of the nameplate available. The use of other spare parts pre- sents potential hazards and will make the war- ranty void.
Page Disposal All device components can be discarded properly after they have been decon- taminated properly. NOTE — Recycling service Kendro Laboratory Products offer, for a small fee, an environmentally compatible recycling service for discarded devices.
Kendro HERAcell 150 Operating Instructions Manual
User Manual - Heraeus - Heracell 150 - Rev. d - 50075549_d_englisch